The 19-channel MPIfR bolometer array which was installed at the telescope in February 1996 will be again available this winter. The horns are located at the center and on the sides of two concentric hexagons with beam spacings . Each channel has a sensitivity of mJy s under good weather conditions and a HPBW of 11''.
The bolometer will normally be operated by wobbling at 2 Hz the secondary mirror in azimuth. As the array if fixed in Nasmyth coordinates, the orientation of the 19 beams with respect to the sky and to the chopping direction changes with elevation. Special software is made available at the telescope for data reduction (NIC  and MOPS ).