STORE PHASE

###
STORE PHASE

            CLIC\STORE PHASE [/BAND LSB|USB|DSB] [/SELF] [/RECEIVER irec]
                                 [/POLAR HORIZONTAL|VERTICAL|BOTH]

        Store the phase calibration determined by the last SOLVE PHASE command.

        One calibration curve is stored for each sideband.  With  option  "/BAND
        code",  one  may  use  the  calibration curve determined with one of the
        sidebands (`code' is UPPER or USB, LOWER or LSB, or AVERAGE or DSB),  to
        calibrate  both  sidebands. With option /POLAR, one may use the calibra-
        tion curve determined with one of the polarizations.

        With option "/SELF", the calibration stored will not be the  calibration
        curve  (as  determined  from the last SOLVE PHASE command) but, for each
        scan, the visibility corresponding to the first subband  set  (eg.,  SET
        SUB  L01 to L03 ...); data from each side band will be used to calibrate
        the same side band. This provide a simple, easy  way  to  self-calibrate
        data taken on strong continuum sources (quasars).

        With  option  "/RECEIVER irec", the calibration curves solved using data
        from receiver "irec" will be stored in parallel with any other  calibra-
        tion  curve.   This phase curve will be selected by giving SET PHASE EX-
        TERNAL (see this command). In practice, this is used for Receiver 2  da-
        ta,  for  which  the  calibration  curve  determined  with Receiver 1 is
        stored. /BAND may be used in conjunction with /RECEIVER.

        STORE PHASE and STORE AMPLITUDE can be  performed  simultaneously  using
        STORE  AMPLITUDE  PHASE  [/BAND code] [/SELF] [/RECEIVER irec] (the /RE-
        CEIVER option has no meaning for amplitudes).