## Functionalities

Each unit can operate in the following modes:

Mode | Band | Channels | Spacing
| (MHz) |  | (MHz)
1 | 320 |  | 2.5
2 | 160 |  | 1.25
3 | 160 |  | 0.625
4 | 80 |  | 0.312
5 | 80 |  | 0.156
6 | 40 |  | 0.078
7 | 20 |  | 0.039

The effective resolution is 1.2 or 1.6 times the channel spacing, depending upon the apodization function.

Each unit can be placed anywhere in the [100,680] MHz IF band, by steps of 0.625 MHz using a local oscillator referred to as the LO3.