RECEIVER /TEMPLOAD
Keywords: pako

Next: RECEIVER /VERTICAL Up: RECEIVER Previous: RECEIVER /SCALE Contents Index

###
RECEIVER /TEMPLOAD

        /TEMPLOAD tempColdLoad tempAmbientLoad
        /TEMPLOAD L[OOKUP]     L[OOKUP]
        /TEMPLOAD *            ...

        Set effective temperatures for the calibration loads at cold and ambient
        temperature.

        Real      :: tempColdLoad      !
        Real      :: tempAmbientLoad   !

        NEW LOGIC FOR SINGLE-PIXEL SIS RECEIVERS (FROM SUMMER 2007, paKo v1.0.7)

        If a numerical value is entered for tempColdLoad or tempAmbientLoad,
        that value is used for calibration calculations.

        Instead of specifying a value one may enter the string L(OOKUP) for
        tempColdLoad and/or tempAmbientLoad.  This is shown in the pakoDisplay
        by the letter "L" instead of a number.

        In this case, during the execution of the observations, the NCS will
        use measured values for the corresponding load temperature(s).

        A * can be substituted for tempColdLoad, which means to leave the
        value for tempColdLoad unchanged from the previous RECEIVER command.

        NOTE: FOR HERA
        measured values are always used during the execution of the
        observations. This is the same logic as in older versions of paKo/NCS.