STORE AMPLITUDE

###
STORE AMPLITUDE

            CLIC\STORE AMPLITUDE [/BAND LSB|USB|DSB] [/SELF]
                            [/POLAR HORIZONTAL|VERTICAL|BOTH]

        Store the amplitude calibration determined by the last SOLVE AMPLITUDE.

        One  calibration  curve is stored for each sideband.  With option "/BAND
        code", one may use the calibration curve  determined  with  one  of  the
        sidebands  (`code' is UPPER or USB, LOWER or LSB, or AVERAGE or DSB), to
        calibrate both sidebands. With option /POLAR, one may use  the  calibra-
        tion curve determined with one of the polarizations.

        With  option "/SELF", the calibration stored will not be the calibration
        curve (as determined from the last SOLVE  AMPLITUDE  command)  but,  for
        each  record,  a  value  derived directly from the data. The solution is
        computed independently for each sideband and each  polarization  by  se-
        lecting  the  matching spectral windows before averaging, so USB/LSB and
        H/V values are stored separately. This provides a simple,  easy  way  to
        self-calibrate data taken on strong continuum sources (quasars).

        STORE  AMPLITUDE  and  STORE PHASE can be performed simultaneously using
        STORE AMPLITUDE PHASE [/BAND code] [/SELF] [/RECEIVER  irec]  (the  /RE-
        CEIVER option has no meaning for amplitudes).